- Western Europe
- Eastern Europe
- Asia and Oceania
- Middle East
The Precision You Need
For The Outcomes You Want.
- Persistent air leaks impact inpatient and outpatient resources utilization, cost, and morbidity. 1,2,3,4
- An air leak present by day 5 should be considered for thoracic surgical consultation or treatment. 2,5,6,7
- A substantial reduction in an air leak using Spiration Valves may accelerate the resolution of an air leak. Complete cessation of an air leak may not be achievable, or necessary, for successful treatment. 6,8,9,10
A proven isolation technique6,12 to identify the source of air leak(s)...
From the leader in valve therapy for air leaks with over two thousand procedures supported worldwide.1. Assess
Block main bronchus to determine if the leak can be stopped or reduced and the length of time it takes to see a change in the water seal monitor.2. Isolate
Systematically work from proximal to distal.3. Place Valve
Once an airway is identified, size the airway and place a valve.4. Reassess
Repeat process to isolate additional leaks as dynamics may have changed since valve placement.*
*Treatment should be limited to no more than 3 segments by placing valves in segmental or sub-segmental bronchi in the target lung to avoid excessive isolation of tissues from ventilation.
Prior to valve placement, it is necessary to isolate and target the culprit airways contributing to the leak. The source and number of air leaks will vary considerably between patients due to changing lung dynamics. It is recommended to begin isolation wtih balloon occlusion at the main bronchus, moving from proximal to distal airways until a culprit leak is isolated.11
Once a valve has been placed, any additional leaks should be located by returning to the main bronchus, and moving from proximal to distal airways. Previously tested airways that showed no evidence of an air leak before a valve was placed may now be visualized in the air leak monitor.
Within a few weeks of the air leak being resolved, the valve(s) may be removed using bronchoscopic techniques and forceps. The unique design of the Spiration Valve has a center removal rod, which facilitates its removal as needed.
Spiration Valve Procedure for Air Leaks
Prior to the procedure, a sizing balloon is calibrated to enable precise measurement of the airways.
Successful isolation requires systematic movement from proximal to distal airways.
The calibrated balloon is used to determine the appropriate valve size for placement.
Valves are removed using bronchoscopic techniques and biopsy forceps.
...for a minimally invasive solution with a center rod to facilitate removal upon resolution of the air leak.
Treatment with the Spiration Valve System has demonstrated a favorable responder rate.9,10,12
- CLICK HERE about the state-of-the-art design of the Spiration Valve System
- CLICK HERE for information on the clinical studies supporting the use of bronchial valves for the treatment of air leaks
- Brunelli. Ann Thorac Surg 2004; 77:1205-1210.
- Schoenenberger. Arch Surg 1991; Vol 126.
- Varela. Eur J Cardiothoracic Surg 2005; 27:329-33.
- Brunelli. Chest 2006; 130:1150-6.
- Cerfolio. Ann Thorac Surg 1998; 66: 1726-1730.
- Mahajan. J Thorac Cardiovasc Surg 2013; 145:626-630.
- Yarmos. Chest 2012; 141(4):1098-105.
- Instructions for use, Humanitarian Use Device (HDE), Spiration Valve System.
- Firlinger. Ann Thorac Surg 2013 Apr; 95(4):1243-9.
- Dooms. Eur Respir J Epub Nov 14, 2013; doi: 10.1183/09031936.00117613.
- Mahajan AK, Doeing DC, Hogarth DK et al. Isolation of persistent leaks and placement of intrabronchial valves. J Thorac Cardiovasc Surg 2013;145:626-30.
- Gillespie. Ann Thorac Surg 2011; 91:270-3.